The Accu-Chek Active blood glucose meter packs many powerful features into a small size. See test result averages up to 90 days, get alerts if you’ve underdosed, re-dose within 10 seconds, set pre- and post-meal reminders to test, and more.Product Manuals
1. Always have a spare set of batteries.
2. Battery life may vary due to factors such as temperature and battery manufacturer.
3. Have a back-up testing method available.
If you have further questions, please contact our Roche Diabetes Care Customer Care line at 080-34-22-38-37 to obtain further support.
Accu-Chek Active (Model GU)
Unique serial number is printed on a label on the device and stored in non-volatile memory
Mutant variant of the glucose dehydrogenase quinone protein (Mut. Q-GDH 2, modified variant of EC 126.96.36.199), acinetobacter spec; photometric end-point measurement
Approximately 5 seconds (blood application with test strip within the meter)
Approximately 8 seconds (blood application with test strip outside the meter)
+8 to +42 °C
Up to 93 % relative humidity
-25 to +70 °C
Up to 500 values with date and time
7, 14, 30, 90 days
Via USB interface (with Micro-USB cable)
New models are pre-coded. Older models require once off installation of a black code chip.
97.8 x 46.8 x 19.1 mm
46 g without battery, 50 g with battery
96 segments LCD display with symbols
Yes (Factory setting ON)
1 battery, type CR 2032
Approximately 1 000 measurements or one year (whichever comes first)
30 or 90 seconds according to operating status
0 to 4,000m (0 to 13,123 feet)
1 to 2 μL
10 mg/dL to 600 mg/dL, 0.6 mmol/L to 33.3 mmol/L
Venous blood anticoagulated with lithium heparin or ammonium heparin or EDTA, arterial blood and blood from neonates
Blood application onto the middle of the test pad. The test strip (and its components) has the function to spread the blood above the test area in a quick and hygienic way
Blood can be applied to an already inserted test strip, or a test strip outside the meter. Outside of the meter dosing is helpful to minimise the risk of contamination when a meter has multiple users (e.g. in a hospital) and, in case of alternative site testing, to facilitate blood application
Additional blood can be added to the test strip within 10 seconds after applying the first drop (applies for inside mode)
Plausibility check option with 5 colour blocks on the test strip vial
20 to 55 % (within-meter mode), 20 to 70 % (outside-meter mode)
18 months after production date. The test strips remain stable at least up to the expiry date printed on the test strip vial, even after opening (the test strip container must be closed tightly after each test strip is removed)
1Evaluation Report, 2013
Testing your blood glucose is an essential part of managing diabetes. The Accu-Chek Active blood glucose meter can help to make it convenient by giving you fast, hassle-free, and accurate results in 4 simple steps.
1. Data on File. Accu-Chek Active Evaluation Report 2013